|Bed Surface Height||450~800mm|
|Back Board Max. Angle||0~80°|
|Calf Board Max. Angle||0~40°|
|Bed Surface Tilt||0~12±1°|
INSTALLATION AND USE
Remarks Of SM3012W
(1) Bed rail descend: grasp the rail and pull the switch forward push down the rail (illustration)
(2)Bed rail elevating: grasp the rail and pull upwards, when the sound of “clatter” is heard, it means the rail has reached the right position.
End Board Installing
When install bed head board and bed end board, firstly turn the metal hook on the clip board upwards, then hitch the clip board onto the hitch nail on the front and back of the bed frame, later lock the metal hook with the hitch nail.
Must not sit at the edge of the bed when the bed is elevating tilting or descending, otherwise the bed board will be distortion due to the uneven strength on the bed.
The bed should use indoor environment.
Before use, please make sure there are no other things under the bed.
Don’t disassemble or detach the products without authorized permission.
Mustn’t use it in the power room or the places full of anesthetic gas or oxygen. But the general oxygen treatment is not forbidden.
STORAGE AND TRANSPORTATION
The hospital bed with packing should be stored in the room ventilated, dry, cool, without heat source and corrosive gas. During transportation, please avoid wet, sunshine and heat source.
MAINTENANCE AND CLEANING
Clean every part of the electric hospital bed frequently.
If the user makes biologic pollution, please use 75％ alcohol to Sterilize and clean it.