The desk centrifuge of low speed series is a machine of beautiful configuration, compactness, lightweight, low noise, a little vibration and easy operation with stable property and convenience of working. It can be used for qualitative analysis in hospital laboratory and biochemical laboratory, and in production of plasma, serum, urea, vaccine, etc.
- Usage Description and Comments
- To the machine is used safety socket. Before use, it must be grounding and put on the stable desk with even surface.
- Firstly, to check the set-time button whether it is showed in position “0”, and then start the source switch, the pilot is twinkling and the power is turning on.
- Put the sample into prepared centrifugal pipe or test tube
- Open the cover and attentively put the centrifugal pipe or test tube with sample into centrifugal shielding tube.
Note: (1) Into the symmetrical shielding tube with the same weight samples for the aims of protecting serious oscillation from unbalance of weight.
(2)Can not put the sample directly into the centrifugal shielding tube.
(3)Put the rubber gasket in the centrifugal shielding tube.
- Close the cover .
- Put the set-time button to required position. If required time less than 5 minutes. At first, set the button more 5 minutes, and then return it to the required set-time position. Put the button to position N.C. “ON” in the case of not use of set-time.
- Regulate the speed button to required value of speed.
- The centrifuge will be auto-stopped after the preset value of time and the set-time button will be returned the position “0”.